- Two kinds of coating thickness measurement principle including magnetic principle and eddy-current principle, available for measuring thickness of the non-magnetic layer covering on magnetic metal and thickness of the non-conductive layer covering on non-magnetic metal.
- High precision mode: multiple measurements and automatically data filtering method to reduce disturbance of measure results
- Temperature compensation: compensate the measurement distortion caused by the drifting of temperature.
- Probe type automatic recognition
|Type of Sensor||F||N|
|Principle||Magnetic induction||Eddy current|
|Measuring range(µm)||0~1500||0~1500 Copper covered with chrome plate(0~40)|
|One point calibration(µm)||±(2%H+1)||± (2%H+1)|
|Two points calibration(µm)||± [(1~2)%H+1]||± [(1~2)%H+1]|
|Minimum radius of curvature(mm)||Bulge1.5||Bulge3|
|Critical Thickness of the base(mm)||0.5||0.3|
There are no reviews yet.