GAOTek Coating Thickness Gauge with User Cal Base (Unit Change) is designed to measure magnetic and non-magnetic materials with an accuracy of ± [(1% ~ 3%). It uses magnetic thickness measurement method to measure the thickness of the magnetic coating on non-magnetic base. This portable device is ideal for paint shops and electroplaters, for office and specialist advisers, for light and heavy engineering, for the aircraft, shipbuilding, automobile and chemical industries.
- Large measurement range: 0 µm-1250 µm.
- Durable material: Wearable ruby probe
- High accuracy: ± [(1%~3%) H+1 µm]
- Easy to use
- Rugged, versatile, one-hand operation
- Auto shut down after 60 seconds of no activity
|Measuring Range||0 µm -1250 µm|
|Accuracy||± [(1% ~ 3%) H +1 µm]
H is thickness
|Auto Shut Off||After 60 seconds (if no activity)|
|Dimensions||4.41 in x 2.72 in x 1.10 in (112 mm x 69 mm x 28 mm)|
|Weight||0.18 lbs. (82 g)|
|Working Temperature||14 °F ~ 140 °F (-10 °C ~ 60 °C)|
- Power-on and Warm-up
Keystroke once and wait until “READY!” is displayed on the LCD. Now the gauge is ready for measurements.
Keystroke twice; when “BASE ZERO” is displayed, place flat on surface; hold steady.
After the measurement is obtained, “DONE” is shown and gauge beeps. Then lift the gauge.
- Two-Point User Calibration
Keystroke three times; gauge enters into the mode of two-point user calibration.
When “USER CAL BASE” is shown on the LCD, please conduct a measurement on the base. When “S1” is shown on the LCD, place Shim S1 on the base and conduct a measurement on Shim S1. When “S2” is shown on the LCD, place Shim S2 on the base and conduct a measurement on Shim S2. When “DONE” is displayed, it is ready for use.
- Unit Change
Keystroke four times; convert from µms to mils or vice versa.
Keystroke five times; gauge resets.
The device powers down after 60 seconds of no activity.