GAOTek Moisture Analyzer for Grain (Fast/High Precision) is used for fast and accurate measurement of moisture in grains such as barley, corn, hay, oats, rape seed, rough rice, sorghum, soybeans and wheat and is useful in the process of allotment, acquisition and storage and machining of packed grains. It saves time and expenses by monitoring dryness and helps to prevent deterioration and decay caused by moisture whilst in storage, therefore making processing more convenient and efficient. It operates by measuring the electrical resistance and comes with automatic temperature compensation feature.
- Portable, compact and easy to use
- Instant moisture measurement readings
- Digital display with back light gives exact and clear readings
- Operates by electrical resistance and has automatic temperature compensation
- Works for a temperature ranging from 14 °F – 140 °F (0 °C -60 °C)
- Measurements can be taken by inserting the sensors into the bale of grains
- Manual Power off at any time and Auto power off after 5 minutes from last operation
|Display||4 digital LCD|
|Measuring range||6 % – 30 %|
|Temperature||14 °F – 140 °F (0 °C – 60 °C)|
|Humidity||5 % RH – 90 % RH|
|Accuracy||±0.5 % n|
|Operation||Electrical resistance method, automatic temperature compensation|
|Power supply||4 x 1.5 AAA size (UM-4) battery|
|Dimensions||18.3 in x 3 in x 1.4 in (460 mm × 75 mm × 35 mm)|
|Weight||0.44 lbs (203 g not including batteries)|
- OPERATION PROCEDURE
1.1 Turn on the power key, the symbol “0” shows up on display. The value must be set to “0”. If the display shows any other value except “0”, press the “0” key. Make sure the probe doesn’t touch anything, else it won’t be efficient. Setting to “0” can decrease the effect of the temperature and humidity in the air.
1.2 Take off the cover of the sensor and insert the probes straight into the materials. The deeper the probes, the more accurate the value.
1.3 The symbol “max” shows on display on releasing the “Hold” key. The max value is stored on display when measuring the process. This function will be cancelled by releasing the “Hold” key again.
1.4 Code choice. Press the “FUN” key and hold it until “CD00” is showed on display,
Press “▲” and “▼” key to choose the code you need, and then release the “FUN” key to confirm it.
1.5 Replace batteries. When the battery symbol shows up on the display, it is time to replace the batteries. Slide the battery’s cover and insert the batteries properly.
- WARNING SETTING
2.1 Press the key 3-9 and hold it until “AL2” shows on display (it takes 5 seconds to complete the operation). Then press “▲” and “▼” key to choose the value you need. Press “FUN” key again to go back to the operation state.
2.2 Setting the “AL1” is the same as “AL2”.
2.3 Usually, “AL2”must larger than “AL1”, but if the “AL2” is less than “AL1” during the setting process, then the instrument will be set back to the factory setting. Ex. AL1=13, AL2=18.
- Replacing batteries.
When the battery symbol shows up on the display, it is time to replace the batteries. Slide the battery’s cover and insert the batteries properly
Please take the batteries out if the instruments aren’t in use for a long time. The probes shouldn’t touch anything while setting the value to “0” or it won’t be efficient.